|Place of Origin:||China|
|Minimum Order Quantity:||5pcs|
|Packaging Details:||4pcs/Carton, 60*30*40cm/Carton|
|Delivery Time:||3-7 working days|
|Payment Terms:||T/T, , MoneyGram, RMB, D/P, D/A, L/C|
|Name:||LK-F17L LED Portable Dental Ultrasonic Scaler With Water Bottle Scaling, Perio, Endo Funtion||Fuse Of Main Unit:||T0.5AL 250V|
|Primary Tip Vibration Excursion:||<90μm||Tip Vibration Frequency:||28kHz+_3kHz|
|Water Entrance Pressure:||0.01MPa~0.5MPa||Function Setting:||G,P,E|
Endo Ultrasonic Scaler Machine,
Perio Ultrasonic Scaler Machine,
Endo portable scaler unit
LK-F17L LED Portable Dental Ultrasonic Scaler with Water Bottle Scaling, Perio, Endo Funtion
|Weight of main unit||1.59kg|
|Rate input||220V 50Hz|
|Fuse of main unit||T0.5AL 250V|
|Primary tip vibration excursion||<90μm|
|Tip vibration frequency||28kHz+_3kHz|
|Water entrance pressure||0.01MPa~0.5MPa|
|Handpiece model||HP-5L with LED lamp|
The portable dental ultrasonic scaler is composed of main unit, handpiece, cable, waterpipe, tip, torque wrench, footswitch, and power supply
The portable dental ultrasonic scaler has following features:
1.1.1 Detachable handpiece can be autoclaved under 134C and 0.22Mpa.
1.1.2 Automatic frequency tracking ensures that the device always works on the best frequency, stable and efficient performance
The portable dental ultrasonic scaler contraindications:
1) The hemophilia patient is forbidden to use this equipment
2) The patients or doctors with heart pacemaker are forbidden to use this equipment.
3)The heart disease patient, pregnant woman and children should be cautious to use the equipment
1.3 Equipment safety classification
1)Operating mode: Continuous operatior
2)Type of protection against electric shock: Class II
3) Degree of protection against electric shock: Type B applied part
4)Applied part of the equipment: Tip
5) Degree of protection against harmful ingress of water: Ordinary equipment
6) Degree of protection against harmful ingress of water: protection degree against water(used on foot switch): IPX
7) Degree of safety of application in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide Equipment can not be used in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide.
The portable dental ultrasonic scaler Cleaning, dlisinfecting and sterilizing:
5.1Please conform to the recommendations of the manual "Reprocessing Instructions of Cleaning, Disinfecting And Sterilizing" delivered with your product regarding the procedure of cleaning, disinfecting, sterilizing and packing of the components.5.2The handpiece (detachable), scaling tips,endochuck, torque wrench, endo wrench,LED lamp and Light conductor (the
handpiece with LED lamp) can be strilized.The handpiece (undetachable), cable and water pipe cannot be sterilized.
a)Clean the handpiece (detachable) with compressed air before sterilization.
b)Be sure that the scaling tip has been unscrewed from the handpiece and it cannot be sterilized with others.
c)Please notice whether the outer of the handpiece is damaged during the treatment and sterilization.Don't smear any protective oil on the surface of handpiece.
d)There are two waterproof "O” rings at the end of the handpiece.Please lubricate them with dental lube frequently, as sterilization and repeated pulling and inserting will reduce their using life.Change a new one once it is damaged or worn excessively.
e) The following sterilizing methods are forbidden:
-Boil in water.
-Dip in iodine, alcohol and glutaraldehyde.
-Bake in oven or microwave oven.
Notice: We are not responsible for any damage caused in the above items.
Contact Person: Mandy Zhang