|Place of Origin:||China|
|Model Number:||LK-J37-2 Endo Smart|
|Minimum Order Quantity:||1pc|
|Packaging Details:||19*13*6cm/pc,G.W. 0.7kg/pc|
|Delivery Time:||3-5 Working Days|
|Payment Terms:||L/C, D/A, D/P, , T/T, MoneyGram, RMB|
750mAh Endodontic Equipment,
1000rpm Endodontic Equipment,
5.0Ncm Root Canal Equipment
Woodpecker Endodontic Equipment Wireless Endo Smart Endo Motor with Contra Angle CE
Endo Smart is mainly used in Endodontic treatment. During root canal preparation procedure, it is used to mold and clean the root canal.
a) Adopt real-time feedback technology and dynamic torque control, effectively preventing needle breakage.
b) Wireless handpiece enables more convenient operation.
c) Wireless charging avoids poor contact problem of traditional contact charging. d) Storage of 9 user-defined modes allows invocation at any time. Under each mode, Continuous Rotation Mode, Reciprocating Motion Mode, and Reverse Rotation Mode are for options
Scope of Application
1 The device is suitable for root canal molding and cleaning in endodontic treatment.
2 The device must be operated in hospital and clinic by the qualified dentists.
Performance and Composition
The device is composed of charging base, handpiece, contra-angle, and power adapter, etc.
The version of software is displayed on the OLED screen when starting up.
Primary technical specifications
1 Battery Lithium battery in handpiece: 3.6V /750mAh
2 Power adapter Input: ~100V-240V 50Hz/60Hz 0.4A Max 5 Output: DC5V/1A
3 Torque: 0.6Ncm-5.0Ncm(6mNm ~ 50mNm)
4 Rotate speed: 100rpm~1000rpm
1.Environment temperature: +5℃ ~ +40℃
2 Relative humidity: 30% ~ 75%
3 Atmospheric pressure: 70kPa ~ 106kPa
1 This device do not include accessories for repair usage, the repair should be carried out by authorized person or authorized after service center. 2 Keep the equipment in a dry storage condition.
3 Do not throw, beat or shock the equipment.
4 Do not smear the equipment with pigments.
5 Calibration is recommended when using a new/other contra-angle or after an extend period of operation, as the running properties can change with usage, cleaning and sterilization.
6 Replace the battery if it seems to be running out of power sooner than it should.
|There is continuous beep sounds after starting the handpiece.||The continuous beep sound is indicating that the handpiece is under reverse rotation state.||Stop the handpiece and change the operating mode to Continuous Rotation Mode.|
|Contra-angle calibration failure||Calibration failure caused by strong resistance of contra-angle||Clean the contra-angle, and recalibrate after oil injection|
a) The bottom of handpiece is heating during wireless charging.
b) Under Reciprocating Motion Mode, the using time is too long
a) Normal phenomenon.
b) Stop use. Use after the temperature of handpiece drops.
|After plugging the handpiece into charging base, the wireless charging indicator does not lights flashing.||a) The handpiece is not in place. b) The handpiece is fully charged.||a) Plug the handpiece in place.|
|The time of endurance becomes shorter after charging.||Battery capacity becomes smaller||Please contact local distributor or manufacturer|
|The continuously rotating file is stuck at the root canal.||Incorrect specification setting. Too high load torque of file.||Choose Reverse Rotation Mode, start the handpiece, and take the file out.|
We also supply many other related products:
Mobile/WhatsApp/Wechat: 0086 18860361012
Contact Person: Echo Chai