|Place of Origin:||China|
|Minimum Order Quantity:||1piece|
|Packaging Details:||1pc/Metal Box, Size:30*28*21cm/Carton Weight:4.5kg|
|Delivery Time:||1-3 working days|
|Payment Terms:||T/T, , MoneyGram, RMB|
|Product:||Portable Dental X Ray||Weight:||4.5kg|
80W Dental X Ray Unit,
7800mAh Dental X Ray Unit,
20KHz Dental X Ray Machine
High frequency DC Dental X Ray Machine Dental Portable Digital X Ray Camera Unit Machine
High frequency dental X-ray machine is used for oral medical institutions taken X-ray diagnosis of the equipment used for lesions in the teeth.
High frequency of dental X-ray machine is mainly composed of high frequency X-ray tube ball head, control circuit, LCD touch display systems, lithium battery power supply system.
The product electric classified as class B type I equipment.
|Power Input Power||100-220VAC/50-60Hz/1.2A|
|Power Output Power||14.8VDC, 10A|
|Li-ion battery capacity||7800mAh|
|Generator Input Power||16.8VAC/2.0A|
|Generator Output powe||80 W|
|Generator Tube voltage / current||60KV/1mA|
|Generator Operating frequency||20KHz|
|Generator Exposure time||0.1-2.5S|
1. High frequency and high resolution image
2. Small volume and light weight
3. LCD touch display
4. Easy maintenance
5. Simple operation
6. Durable use
7. Toshiba tube
Control system description
1. The microprocessor control system.
2. According to the tooth bit and mode automatically choose the exposure time , time range: 0.1 -2.5 S.
3. Manually adjust the exposure time, every time to adjust time of 0.1 S.
4. Equipped with three block mode selection switch, respectively is: children, adults and RVG (digital sensing mode).
5 .Manual switch can choose can extend and hand control wire.
1.Into the normal state, the next step selection mode; There are three kinds of modes, respectively, children and adults RVG (digital sensing mode); Selection mode is completed, the next step to choose to shoot tooth area, respectively have, In the incisor, premolar and molar teeth after a regional choice. Every area of the teeth have a shot time please refer to the following table data: (Note: Because of different sensors and film sensitivity is different；Please adjust the exposure time before using,the table below data is for reference only.)
2.The pattern and the tooth to adjust well, the next step into the shooting Angle adjustment phase, different teeth corresponding to different angles, detailed parameters refer to see table data:
Transport and storage conditions
This equipment in stored in the transport process, can not be stored in the corrosive gas, chemicals and flammable and explosive environment, in the process of transport do not drag, roll, throwing, extrusion, etc all strenuous exercise.
1. Ambient temperature: -20℃-55℃
2. Relative humidity: 10%- 60%
3. Atmospheric pressure: 500Hpa-1060Hpa
Disposal of scrap equipment
This equipment is a high frequency of dental X-ray machine, it is made of a variety of materials, including iron, copper, lead and other metals, plastics, electronic components, insulating oil, X ray tube, etc.; When the system is discarded, ban thrown in the natural environment, must be in accordance with the relevant laws and regulations to the professional recycling company for processing.
Fault code display
|Fault Code||Fault Description||Corresponding Action||Solution|
|The battery voltage is too low||Turn off the power||Immediately charging is completed, will be used regain.|
|Z02||Filament voltage is too low||Turn off the power||Immediately charging is completed, will be used regain.|
|Z03||Output over-current protection||
Turn off the power
Exposure to ban
|Turn off the power, Wait for 15 minutes and then use.|
|Z04||X- ray tube temperature is too high||
Turn off the power
Exposure to ban
|Turn off the power, Wait for 30minutes and then use.|
Besides, we also have many other related dental X-Ray items:
Contact Person: Echo Chai