|Place of Origin:||China|
|Minimum Order Quantity:||5pcs|
|Packaging Details:||5pcs/carton 60*40*33cm G.W.2.7KG|
|Delivery Time:||500pcs per month|
|Payment Terms:||TT, , Money Gram, RMB|
|Power Source:||220V±10% / 50Hz ± 1Hz 150mA||Ain Unit Input:||24V 50Hz ± 1Hz 1.3A|
LED Dental Ultrasonic Scaler,
Scaling Dental Ultrasonic Scaler,
Perio piezo ultrasonic scaler
Colorful LED Dental Ultrasonic Scaler with Scaling and Perio Function
Automatic frequency tracking
Accurate feedback control and constant power output, ensure a more comforatble feeling to patients.
Power source:～220V±10% / 50Hz ± 1Hz 150mA
Main unit input:～24V 50Hz ± 1Hz 1.3A
Main unit weight:0.75kg
Power supply weight:1kg
Tips: G1*2, G2, G4, P1
Trouble Shooting to Ultrasonic Scaler
1 The ultrasonic scaler worked fine before. But recently, power decreased obviously. What is the reason?
First, check if ther is water or stains in the connection between handpiece and cable. The water or stains will cause electricity leakage, generate heat, decrease tip vibration, even stop water atomizing properly. So, it is very important to make sure cable and scaler dry before usage.
Solution: Dry handpiece and cable connector. Moist the two O rings of the connector with a drop of clean water, then connect. Power should be recovered mostly.
2 Handpiece cable connector is quite clean. But the power is still very weak. What is the reason?
First, check if the tip is well screwed into the handpiece
If the tip is not connecting well, vibration cannot be passed to it. Even you switch the power to max, tip is still very weak. Atomizing is bad. Heat will also be generated because of rub from handpiece and tip.
Solution: Screw the tip tightly. Before you screw the tip, you need to unscrew it and clean the dirt around screws with alcohol cotton. The dirt will also make tip loose. After that, screw the tip tightly. Power will be recovered.
3 Why does the scaler tip become loose?
The tip is vibrating during working. It will be loose after a long time vibrating. In fact, being loose is normal phenomenon.
4 Why my scaler tips always loosen? Sometimes it can’t even treat one patient!
The tip is not screwed tightly. Generally there are two reasons:
One: there is dirt between the screw thread of tips and handpiece.
Two: Tip is not screwed tightly enough. You should screw it more with torque wrench.
5 How do I know I use proper power to screw it?
Torque wrench should be used to screw the tips.
6 While handpiece connector is clean, and scaler tip is screwed tightly,
however, tip vibration is still very weak. And water spray is weak, too. Why?
A. Scaler tip might be cracked. Generally, new scaler tip has good performance. So it works fine even if you switch the power into low stage. If you switch the power to higher stage when you use new tip, tip get worn very fast. So please switch to low power stage when you use a new tip. As the tip get worn slowly, just increase power accordingly. In this way, the tip can be used longer while operators will not feel tip vibration get weaker.
B. Scaler tip is worn too much. As scaler tip gets worn gradually, power stage has to be increased accordingly. After the power stage is increased to max, the tip will get worn further, and there is no way to compensate the vibration further. Vibration will be weaker and weaker. The final and easy solution is to change a new tip.
7 Both vibration and water spray are good, but you can feel scaler tip is heat when you touch it. Is this a problem?
No, it is not a problem. As the scaler is driven by ultrasound, the tip vibrates, which generate hole effect. Hole effect from vibration will clean stains and calculus fast. It is noticed that the calculus is not scaled by scaler, but moved by vibration. Nerves of gingival, lips, fingers etc can feel heat or hurt during vibration. That is the reason. Generally, it will not burn people. Heat from finger proves that scaler tip’s power, not problem.
8 The machine works fine. But scaler handpiece connector is full of rust. Will this bring trouble to machine itself?
Yes. This connector is to transfer water and electricity. Little rust will not make any influence. Too much rust will not only affect circulation of electricity, but also cause electricity leakage, even burn the whole circuit.
9 How to avoid the rusting of the scaler handpiece connector?
A: Make the scaler handpiece connector dry with 3 way syringe. Then put one drop of water into the rubber O ring.
B: Please make sure waterproof cover buckled to prevent the disinfectant corrode metal parts when the handpiece are disinfecting.
10 The scaling tip doesn’t vibrate but there is water flowing out when stepping on the switch. And sometimes the tips will not vibrate until you touch it.
Solution: Please use compatible scaler tips.
Contact Person: Chloe Yao
Surgical Dental Diamond Burs Tooth Polishing FG Type High Speed Handpiece Support
210pcs Sample Bur Book FG Diamond Head With Calculator Exhibition Dentist Supply
Disposable Dental Abrasive Polishing Natural Rubber Material Prophy Cup RA 100pcs/box
Opening Type Dental Abrasive Bur Disinfection Box 30 holes Colorful FG RA
Led Light Professional Laser Teeth Whitening Machine Blue color Wheeled Easy Install
Zoom Teeth Whitening Unit Red Color Lighting 300W 40-50cm Clamping Poweful
Dental Teeth Whitening Led Lamp Cold Light 18V Irradiation Two Or Three Times Each Course
35% Hydrogen Peroxide Whitening Gel , At Home Teeth Whitening Gel Dental Professional Strength
Protaper Dental Gutta Percha Points , Dental Absorbent Paper Points F1 F2 F3 Universal
GP Protaper Paper Points For Dentistry Gutta Percha Dental Filling Millimeter Marked
Protaper Gutta Percha Root Canal Filling Material Rubber Based Dentistry Support
Like Waveone Dental Gutta Percha Points , Protaper Gp Points Dental F1 F2 F3 Sizes Available