|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Packaging Details:||1piece/Metal Case, carton size: 57*45*22, G.W.: 8.9kg|
|Delivery Time:||2-3 working days|
|Payment Terms:||T/T, , MoneyGram, RMB, D/A, D/P|
|Torque Range:||5.0-60N.cm(20:1)||Motor Rotation Speed Range:||500-40000rpm(1:1)|
|Coolant Flow:||0-100ml/min||Ouput Power:||210W|
210W Dental Implant Surgery Motor,
100ml/Min Dental Implant Surgery Motor,
40000rpm Dental Implant Machine
1) Intuitive LCD display and control panel, which bring convenience to the operation;
2) Original imported brushless motor, which is characterized by great torque, low noise, non-vibrations as well as accuracy and can be sterilized by high temperature;
3) Alternative input supply voltage: AC100-110V or AC220-240V;
4) Multifunctional foot switch, which is not only handy to operate but also against cross-infection caused by the contact between hands and the device;
5) Adjustable rotation speed, torque, coolant flow and forward/reverse, which can meet various demands during implanting process;
6) Ten types of reduction control angles, which are able to be selected as you need for operations;
7) Ten kinds of program settings, each of which allows setting up different numerical values of rotation speed, torque, coolant flow, forward/reverse or reduction ratio for practical needs.
|Input Power Supply||AC100-110V or AC220-240V 50/60HZ|
|Motor Rotation Speed Range||1000rpm-30000rpm (1:1)|
|Torque Range||5.0-55N.cm (20:1)|
|Stored Programs||10 types|
|Foot Switch Control||coolant flow, program forward/reverse, speed|
All standard operation can be completed by foot control
Turn on the main switch. Control panel is ready to execute the program 1 stored
in the programs.
Select the desired program number. Push the PROG button by foot, program
numbers will always ascend to the next. Selected program number displays on
the LCD screen and it will roll from No.10 program onto No.1 program.
Verification program details. Verify the details of displayed programs.
Maximum number displayed shows the set up of the speed.
Run the micro-motor. Step on the central part of the speed pedal of the foot
control, the micro-motor starts running. If the coolant flow is also incorporated
into the program, the cooling pump will run automatically. The pedal is
stepped more heavily, the speed will be faster. When the pedal is stepped on
the bottom, speed reaches the set maximum value.
Torque limit. In the running of the micro-motor, if the load on moving of the
drilling reaches the maximum torque, then the micro-motor will stop working
automatically. Re-push the foot pedal to let the motor start working again.
Stop the micro-motor. Release the foot control, micro-motor will stop
Change the rotation direction of the motor. Change the rotation direction of
the micro-motor (and the drill), just push the Forward/ Reverse button. When
the rotation direction is under reverse mode, warning beep sound can be
There are also many other related items:
Contact Person: Mandy Zhang