|Place of Origin:||China|
|Minimum Order Quantity:||1piece|
|Packaging Details:||Carton, weight: 4kg, Size: 33*30*33cm|
|Delivery Time:||1-3 working days.|
|Payment Terms:||T/T, , MoneyGram, RMB, D/A, D/P, L/C|
|Exposure Time Range:||0.01-1.6sec||Maximum Duty Cycle:||1:60|
|Maximum Inherent Filtration:||1.6mm Al||Maximum Output Power:||120W Nominal At 60KV,2mA|
|Generator Rating:||2mA At 60KV±5%||Leakage Technique Factors:||60KV,2mA,1.6sec|
60KV Dental X Ray Machine,
Battery Dental X Ray Machine,
60KV handheld x ray dental
Korea Design 60KV Dental X Ray Machine with Rechargeable Lithium Polymer Battery
The time setting delivers optimal results to each of digital sensor or film-based images which are done on a
Please note that digital imaging sensors vary greatly in their characteristics and they could require significantly
diferent exposure settings than the ones given by the chart table for exposure time below.
The fllowing time table is a recommended nominal exposure condition.
Please adjust the device to meet the operator.Os density preference.
It is recommended to make annual maintenance checks by a qualifed technician for performance and safety as
well as assurance of accurate X-ray exposure levels.
|Rechargeable Lithium Polymer battery||24V|
|Maximum battery charge||25.2V|
|Battery current at 2mA,60KV output||10A|
|Power supply||JBL7451250750100FJ,Bridge Power Corp.|
Contact Person: Jennifer Su