|Place of Origin:||China|
|Model Number:||LK-J37-2 Endo Smart|
|Minimum Order Quantity:||1pc|
|Packaging Details:||19*13*6cm/pc,G.W. 0.7kg/pc|
|Delivery Time:||1-3 Working Days|
|Payment Terms:||L/C, D/A, D/P, Western Union, T/T, MoneyGram, RMB|
|Type:||Wireless||Torque:||0.6Ncm-5.0Ncm(6mNm ~ 50mNm)|
|Power Adapter:||100V-240V 50Hz/60Hz 0.4A Max||Battery:||Chargeable|
Oled Screen Endodontic Equipment,
Oled Screen Woodpecker Endo,
Wireless Chargeable Woodpecker Endo
Wireless Oled Screen Endodontic Equipment Woodpecker Endo Smart Endo Motor
a) Adopt real-time feedback technology and dynamic torque control, effectively preventing needle breakage.
b) Wireless handpiece enables more convenient operation.
c) Wireless charging avoids poor contact problem of traditional contact charging. d) Storage of 9 user-defined modes allows invocation at any time. Under each mode, Continuous Rotation Mode, Reciprocating Motion Mode, and Reverse Rotation Mode are for options
Installation and removal of contra-angleInstallation
Align the positioning pin of contra-angle with the positioning hole
of handpiece, horizontally pushing the contra-angle. A click
indicates that it is well installed. By aligning those three pins on contra-
angle with those six holes on handpiece, the contra-angle can be
installed in different angle. (As shown below)
Pull out the contra- angle horizontally when the motor handpiece does
a) Before plugging in or pulling out contra-angle, please first stop the
b) After installation, please check and confirm that the contra-angle has
been well installed.
Primary technical specifications
1 Battery Lithium battery in handpiece: 3.6V /750mAh
2 Power adapter Input: ~100V-240V 50Hz/60Hz 0.4A Max 5 Output: DC5V/1A
3 Torque: 0.6Ncm-5.0Ncm(6mNm ~ 50mNm)
4 Rotate speed: 100rpm~1000rpm
1.Environment temperature: +5℃ ~ +40℃
2 Relative humidity: 30% ~ 75%
3 Atmospheric pressure: 70kPa ~ 106kPa
Customized program handover setting
In Standby interface, shortly press adjusting key“+”and adjusting key
to realize handovers among different customized programs. Long
press adjusting key“+”and adjusting key“”to realize cyclic handover.
Contra-angle calibration setting
After replacement of contra-angle, the contra-angle shall be calibrated
before use. In Standby Interface, first long press setting key“S”
then long press“”for 2s to enter Calibration Interface of contra-angle.
After 15s's countdown, the interface of successful calibration will
|There is continuous beep sounds after starting the handpiece.||The continuous beep sound is indicating that the handpiece is under reverse rotation state.||Stop the handpiece and change the operating mode to Continuous Rotation Mode.|
|Contra-angle calibration failure||Calibration failure caused by strong resistance of contra-angle||Clean the contra-angle, and recalibrate after oil injection|
a) The bottom of handpiece is heating during wireless charging.
b) Under Reciprocating Motion Mode, the using time is too long
a) Normal phenomenon.
b) Stop use. Use after the temperature of handpiece drops.
|After plugging the handpiece into charging base, the wireless charging indicator does not lights flashing.||a) The handpiece is not in place. b) The handpiece is fully charged.||a) Plug the handpiece in place.|
|The time of endurance becomes shorter after charging.||Battery capacity becomes smaller||Please contact local distributor or manufacturer|
|The continuously rotating file is stuck at the root canal.||Incorrect specification setting. Too high load torque of file.||Choose Reverse Rotation Mode, start the handpiece, and take the file out.|
We also supply many other related products:
Mobile/WhatsApp/Wechat: 0086 18860361012
Contact Person: Daisy Dai