|Place of Origin:||China|
|Certification:||CE, ISO, FDA|
|Minimum Order Quantity:||5pcs|
|Packaging Details:||20pcs/carton, G.W: 12kg,50*30*40/carton|
|Delivery Time:||2-3 working days|
|Payment Terms:||TT, , Money Gram, RMB, L/C, D/A, D/P|
|Product Name:||Woodpecker Built In Dental Chair Type Portable Dental Ultrasonic Scaler With HW-3H Detachable Handpiece||Feature:||Built In To Dental Chair|
|Scaler Tips:||G1*2, G4, G2, P1||Detachable Handpiece Model:||HW-3H|
|Compatiable With:||EMS||Water Pressure:||0.01MPa To 0. 5MPa|
0.5MPa Dental Ultrasonic Scaler,
Handpiece Dental Ultrasonic Scaler,
0.5MPa portable ultrasonic scaler
Woodpecker Built in Dental Chair Type Portable Dental Ultrasonic Scaler with HW-3H Detachable Handpiece
The Portable Dental Ultrasonic Scaler Usage:
UDS-N2 is used along with dental unit for teeth cleaning.They are also indispensable equipments for tooth disease prevention and treatment.
Portable Dental Ultrasonic Scaler Components
The components of the machine are listed in the packing list.
Components and scope of application
a) Portable Dental Ultrasonic Scaler is composed of elect circuit,water way and ultrasonic transducer.
b) Portable Dental Ultrasonic Scaler is used for the dental calculus elimination and root canal treatment.
The Portable Dental Ultrasonic Scaler main technical specifications
|Power input With transformer||220-240V~50Hz/60Hz 150mA|
|Power input Without transformer||24V~50Hz/60Hz 1.3A|
|Output primary tip Vibration excursion||=100um|
|Output half-excursion force||=2N|
|Output tip Vibration frequency||28kHz士3kHz|
|Output power||3W to 20W|
|Water pressure||0.01MPa to 0.5MPa|
|Weight of main unit||0.2kg|
|Weight of transformer||1kg (optional)|
|Operating mode||Continuous operation|
Type of protection against electric shock
Degree of protection against electric shock: Type BF applied part
Applied part of the equipment: handpiece and tip
Degree of protection against harmful ingress of water: Ordinary equipment
Degree of protection against harmful ingress of water: protection degree against water (used on foot switch): IPX1
Degree of safety of application in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide: Equipment can not be used in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide.
The Portable Dental Ultrasonic Scaler Working condition:
Environment temperature: 5'C to 40C
Atmospheric pressure: 70kPa to 106kPa
The Portable Dental Ultrasonic Scaler Storage and maintenance
The equipment should be handled carefully and lightly.
Be sure that it is far from the vibration, and installed or kept in a cool, dry and ventilated place.
Don't store the machine together with the articles that is combustible, poisonous, caustic, orexplosive.
This equipment should be stored in a room where the relative humidity is 80%, atmospheric pressure 50kPa to 106kPa, and the temperature-10C~+50C.
Please turn off the electrical source if not be use it. if not use for a long time, please make the machine get through to the power and water once per month for five minutes.
Contact Person: Mandy Zhang