|Place of Origin:||China|
|Certification:||CE ISO FDA|
|Minimum Order Quantity:||1pc|
|Delivery Time:||1 day After Got Payment|
|Payment Terms:||L/C, D/A, D/P, T/T, , MoneyGram|
|Package:||Metal Handle Box||Tube Voltage:||60kv|
|Place Of Origin:||China||Warranty:||1 Year|
RVG Dental X Ray Machine,
60KV Dental X Ray Machine,
0.1MA Imaging X Ray Machine
Portable RVG Digital Dental X Ray Machine High Frequency 60KV Imaging X ray Machine
The model combines the advantages of similar products at home and abroad, improves the shortcomings of the on-frequency X-ray machine: such as the great current intensity, great amount of junk-ray, etc.. The tube voltage frequency of this machine is 30KHz, and the tube current is 0.1MA. The radiation scope is within an included angle of 24 degrees, within a distance of 1.2 meters ahead. This is pioneering in China as well as in abroad, and is named the "green X-ray machine."
|Properties||Medical X-ray Equipments & Accessories|
|Kind of power||Electric|
|After-sales service||Online Technical Support|
Develop the Dental film
1. Picture the tooth following the right operating procedure of the X-ray Unit, then take the film out from the mouth.
2. Use a syringe to extract 4-5 ml imaging liquid
3. Transfuse the liquid into the bag, nip the both ends and sides of the bag with thumbs and first fingers(knead the film).
4. Open the end of the bag about ten minutes after transfusing the liquor into the bag
5. take the film out and put it into clear water and rinse it. Don’t wipe the surface of the film with fingers in case the image will be damaged.
6. Observe and diagnose after the film becomes dry.
Points for Attention
1. Make sure the angle of film, tooth and ball head should be proper when taking a picture, keep steady as the last step of picturing, make sure there is no position changing of the three.
2. Turn the power off when work is done.
3. Change the battery if the remote controller lose affection or remote distance is shorted.
4. Take pictures one minute later after it’s turned on, to keep the ray output steady.
5. The equipment turns into protection mode automatically when the voltage is wrong. Then it will not be able to carry out the normal work of filming.
6. When taking pictures, the ON/OFF button could stop the objection and then the machine goes back to the preparation mode.
7. Use high quality tooth film and developing liquid to make clear pictures.
8. Hand the ball head gently whening using, so as not to damage that precious possession of the ball.
9. Do not pull hard on the cable which connects the ball head and main unit.
10. The cable connecting the ball head and the main unit cannot bear the weight of the ball head, so do not raise the ball head by holding the cable, it will cause damage.
11. We usually set the time between 3-8 seconds when taking picture, but the actual time of ray output is 0.2-0.5 seconds.
12. For any situation that the machine cannot take regular pictures, contact the seller to solve the problem instead of opening the unit by yourself.
13. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.
14. Developing of films should be operated at temperatures between 23-25 degrees to ensure the image quality.
15. Other requests in accordance with the scope of the technical parameters.
16. Please read the manual carefully before using.
Contact Person: Daisy Dai