|Place of Origin:||Guilin, China|
|Minimum Order Quantity:||5pcs|
|Packaging Details:||Carton, Gross Weight: 145g, Dimensions: 260*31*34mm|
|Delivery Time:||3-7 working days|
|Payment Terms:||T/T, , MoneyGram, RMB, D/A, D/P, L/C|
|Name:||LK-G32 1s Broad Spectrum LED Dental Light Cure Unit Lamp 1 Sec Curing Long Battery Life 10sec*1300times||Power Input:||100V-240VAC 50Hz/60Hz|
|Power Output:||DC 5V/1A||Wave Length:||420nm-480nm|
|Light Intensity:||1000m/cm2 - 2500mW/cm2||Battery:||2500mA|
1s Dental Light Cure Unit,
Spectrum Dental Light Cure Unit,
1s led curing light dental
LK-G32 1s Broad Spectrum LED Dental Light Cure Unit Lamp 1 sec Curing Long Battery Life 10sec*1300times
The LED Dental Light Cure Unit suppose to be used in hospital and dental clinic, should be used by trained qualified dentist. This Curing Light is used for the principle of ray radiation to solidify the light-sensitive resin by shooting at it in a short time, it is used to restore teeth and solidify material for whitening teeth.
The LED Dental Light Cure Unit Technical Datas:
|Power Input||100V-240VAC 50Hz/60Hz|
|Power Output||DC 5V/1A|
|Light intensity||1000m/cm2 _ 2500mW/cm2|
The LED Dental Light Cure Unit Operation:
4.1 Modes setting: Lightly press the mode key to change the running modes.The Curing Light will shine the diffrent llluminated blue light.The available modes see the Technical Parameters table.
4.2 Time setting:Lightly press the time button to choose the solidification time.The available time setting see the Technical Parameters table.
4.3 During the operation, aim blue light at the position needing solidification.Press the power button "ON/ OFF" switch, a "beep" sound will appear, the Curing Light starts to work under the selected mode.Then it counts down to "o" second to end the solidification.
4.4 During operation, the blue light can be stopped by press the power button "ON/OFF" at any time.
4.5Low power detect circuit is fixed inside of the main unit,when low power is detected, the indicator of the
main unit will wink, please charge in time.
4.6 After the operation,please clean the fiber with calico in order not to affect the light intensity.
4.7 This equiprment will turm off automatically if no any action within 2 minutes, turn it on by press "ONOFF" button.
4.8 The depth of solidification of composite is no less than 4mm per 10 seconds.
4.9 The Curing Light is equipped with over-heat protection system. It can continuously work 200s,For example,continuously operate the curing light for 10 times under 20s working mode, then it will come into over-heat protection status.And only after 2 minutes sleep, it can restart working 200s continuously.
The LED Dental Light Cure Unit Cleaning,disinfecting and sterilizing:
5.1 Please conform to the recommendations of the manual "Reprocessing Instructions of Cleaning,
Disinfecting And Sterilizing" delivered with your product regarding the procedure of cleaning, disinfecting,
sterilizing and packing of the components.Follow present-day regulations enforced in the country about
reprocessing.The use of scouring powder or an abrasive sponge will damage its surface.
5.2 The optical fiber can be sterilized.The light hood,LED lamp, battery and power adapter cannot be sterilized.
Contact Person: Mandy Zhang