|Place of Origin:||China|
|Minimum Order Quantity:||1piece|
|Packaging Details:||147*110*122cm/Wooden Case 40HQ:32pcs|
|Delivery Time:||3-7 working days|
|Payment Terms:||L/C, D/A, D/P, T/T, , MoneyGram.RMB|
|Supply Ability:||1000pcs per month|
|Supply Voltage:||220V/110V||Instrument Tray:||Touch Screen|
|Memory Position:||9||Lamp:||Implant LED Sensor|
|X-Ray Film Viewer:||LED Panoramic||Dentist Stool:||1pc Metal For Free|
220V Dental Chair Equipment,
440mm Dental Chair Unit
Implant LED Sensor Lamp LK-A14TD Dental Chair Equipment Touch Screen 9 Memory
8 colors optional
Luxurious Implant LED sensor lamp
Spirit lock tight Equilibrium Equip
Compensate Dental Unit
Multi-function switch feet steps
King Size Cushion
9 Memory Position
Strong and weak operation light
Four-way high speed air turbine H.P. 2sets
Powerful suction apparatus 1set each
Four-way low speed air motor H.P. 1set
Water suction and saliva ejector 1set
Unify move glass spittoon
Three way springe 2sets
Arm Rest 2pcs
Dental stool 1set
Lay in water of insides pot
Touch Screen Display Instrument tray with air brake
Supply Voltage: 220V 50Hz
Air Pressure: Larger than0.55Mpa
Water Pressure: 0.2Mpa-.04Mpa
Lowest position of chair: 440mm
Tallest position of chair: 860mm
Chair-back highest 5°
Chair-hack lowest 85°
|Lamp||Implant Sensor LED|
Curing light 1set
Ultrasonic Scaler 1set
Intra-oral Camera 1set
Electric Micromotor 1set
Fiber Optic System 2tubes
40hq container 32pcs
It’s necessary to maintain the chair regularly, in order to keep the chair wear well and beauty always.
(1) The chair will be ageing by touching with oil and the other corrosive, it should be avoided. The pillow can be taken down to disinfected with medical alcohol also.
(2) The surface can be cleaned and disinfected with medical alcohol to keep it beauty, at the same time, not to touch it by sharp or something hard.
(3) When you didn’t use the chair, please down the chair to the lowest position and up about 5cm. Stay the chair in dry environment that you can use it for a long time.
Clean and disinfection:
Please refer to the instruction for the disinfection of H.P.
2. Disinfection of the suction pipeline
Sterilize the suction pipeline in the high-pressure, high-temperature, vacuum, and disinfection box, suggested to use the throwaway suction pipeline.
Weak suction tubes interface specification: Ф7mm
Strong suction tubes interface specification: Ф11mm
3. Disinfection of the spittoon
Use running water to wash the spittoon first, use alcohol after is OK.
4. Disinfection of the three-way syringe
Press the button of the three way syringe and take down the nozzle, then Sterilize
the nozzle in the high-pressure, high-temperature, vacuum, and disinfection box,
put it back to the rack when it finish disinfection.
Spare parts replacement:
1. Protector tube replacement
Have assembled a fusible cut-out for avoiding causing bad effects on consumer's current supply network because of this machine short circuit malfunction on this machine circuit, the fusible cut-out assembles within dental chair front cover , the step is in progress time replacement according to the following:
1. How to place the order?
Please send inquiry to us. We will help you confirm the products and the quantity you need, and send you
the invoice and payment information.
We can accept the payment by T/T, Paypal and .
2. Can you provide samples? Are the samples free?
For small products, such as endo files, dental burs, acrylic resin teeth, retractors, and other small materials,
we can provide free sample, but the buyer need to pay the freight in advance.
3. How about the warranty?
For non-consumable products, like dental chair, X ray unit, ultrasonic scaler etc, we can provide one year warranty.
Please provide photo or video to describe the problems, we will suggest solutions or send spare part. Shipping cost
will be undertaken by the buyer.
4. Do you have CE or FDA for your products?
Yes, most products have CE and FDA certificates. Our products have been exported to many countries successfully.
Contact Person: Mandy Zhang