|Place of Origin:||China|
|Minimum Order Quantity:||Two sets|
|Delivery Time:||1-3 working days|
|Payment Terms:||L/C, D/A, D/P, T/T, , MoneyGram, RMB|
|Supply Ability:||2,000pcs per month|
|Heating Time:||2seconds||Working Temperature:||150 C/200 C|
|Output:||DC5v, 2.4v||Adpter Input:||100v-240v AC|
|Feature:||Short Heating Time,up To 150 C Within 2s||Title:||Double Batteries Gutta Percha Dental Obturation Pen With Short Heating Time|
root canal treatment equipment,
dental apex locator
Double Batteries Gutta Percha Dental Obturation Pen with Short Heating Time
The obturation pen support the vertical compaction of Gutta-percha for the root canal (mainly is the apical parts) filling. It is intended exclusively for use by trained dentist only in clinic or Laboratory.
|Power Adapter||Input: 100~240V/AC 50/60Hz
Output: DC 5V/1.5A
|Li-ion Battery||Voltage: DC3.7V
|Temperature of Pen Tip||Low temperature status: 160~200℃
High temperature status: 200~240℃
1.If the unit is under low voltage, all the Led indicator will flash at
the same time, and send out alarm, then the unit will
automatically shut down, and the battery should be charged by
the charging base.
2.The charging base can be available for one batteries .
When it connect to the power source, the LED
indicator on the adaptor will be turned on.
3.In the process of charging, the indicators on charging base will
display green; After finishing the charge, it will turn green and
the system will automatically stop charging.
4.The unit should be continuous charging more than 2 hours
from low voltage to full charge
1.To avoid the overheating, the device will stop and into the
protection mode automatically after heating over 30s. To
continue the heating, release the button during the protection
mode and press the button again.
2. For saving the power, the unit will be turned off after 2mins
without any operation
Mobile/WhatsApp/Wechat: 0086 13663715484
Contact Person: admin