|Place of Origin:||China|
|Certification:||CE ISO FDA|
|Minimum Order Quantity:||1pc|
|Delivery Time:||1 day After Got Payment|
|Payment Terms:||L/C, D/A, D/P, T/T, , MoneyGram|
|Package:||Metal Handle Box||Tube Voltage:||70kv|
|Color:||Blue, Red, White||Warranty:||1 Year|
70KV Dental X Ray Machine,
80KHz Dental X Ray Machine,
0.3MM Orthodontic X Ray Machine
Korea Desgin Dental X Ray Machine Colorful High Frequency 70KV Rayme
Tube Voltage 70Kv
Tube current 1MA
Focus points 0.3MM
Time of Exposure 0.5-9s (0.05-1.2s)
Radiation Leakage less than 20μgy/h
Rated Power 60W
|Power requirement||Battery Charger|
|Exposure Time||0.5-9s (0.05-1.2s)|
|After-sales service||Online Technical Support|
a) The rated parameters of x-ray production device
◆Tube voltage: 70kv (fixed)
◆Tube current: 1MA (fixed)
◆High voltage generating circuit: High frequency switch
◆X-ray control method: Microprocessor control
◆Time setting range: 0.05-1.2 seconds
◆Type: bipolar fixed x-ray tube
◆X-ray tube Model: DG-073B-DC
◆The size of x-ray tube focal spot: 0.3*0.3mm
◆ANODE heating unit: 8.5Khu
◆Maximum tube voltage: 75Kv
◆Target angle: 21°
◆Total filtration: 1.6mmAL (Origin filtration: 0.8mmAL, Additional filtration: 1mmAL)
◆X-ray irradiation range: 73mm (roundness)
e) Useage voltage: battery DC16.8V
f) Charger: Input AC100-240V, Output: DC16.8V
Points for Attention
1. Make sure the angle of film,tooth,and ball head are properly aligned when taking a picture. Keep them steady until the last step of picturing; making sure that there is no change in any of their positions.
2. Remember to turn the power 0ff when work is done.
3. The equipment should not be repeated taking photos within three minutes.
4. The battery DC socket connected to the DC IN socket on the back of the machine, and confirmed that the back of the charging LED is red
5. Battery charging will be used machine’s accessories dedicated charger.
6. The battery is a consumable item, with the passage of time; the time of usage will be slowly reduced compared with the initial purchase. For safe use, it is recommended to replace the battery 2 years
7. The battery is the high efficiency of the lithium-polymer battery. When charging for one time, X-ray may be irradiated more than 100 times.
8. To reach accurate the amount of X-ray irradiation, it is recommended to use good sensitivity film and sensors which is level 4. According to the different requirments of manufacturing company, it requires the user to change the irradiation time
9. The equipment switches into protection mode automatically when the voltage is incorrect. If this occurs, it will not be able to carry out the normal work of filming.
10. Handle the ball head gently while in use,so as not to damage the delicate component.
11. If a situation occurs, preventing the machine from taking regular pictures, contact the seller to solve the problem instead of trying to service the product yourself.
12. Keep the tooth film and the liquid in a proper place and use them within the shelf-life period.
13. Other requests accordance with the scope of the technical parameters.
14. Please read the manual carefully before using.
Linker Dental Group Limited
Contact Person: Daisy Dai