|Place of Origin:||China|
|Certification:||CE, ISO, FDA|
|Minimum Order Quantity:||10|
|Delivery Time:||2-3 working days|
|Payment Terms:||T/T, , MoneyGram, RMB|
|Diameter:||2.35mm / 1.6mm||Usage:||Dental Polisher|
|Materila:||Aluminum Alloy||Connector:||FG / RA|
|Name:||Contra Angle Handpiece||Speed:||High Speed / Low Speed|
air turbine handpiece,
air powered dental drill
Push Button Dental Turbine Handpiec Contra Angle RA 2.35mm / FG 1.6mm
Push button type: available
Package size: 15*7*3cm
Chuck type: Push Button
Contra Anlge Hand Piece Fit HP Dia.2.35mm / 1.6mm Pins/Burs
Means For Sterilizing:Prevacuumizing ,Autoclave <135℃
Latch Type Contra Angle Fit RA/CA Dia.2.35mm Burs Air Motor 4-Holes or 2-Holes Coupling Rated
or Handpiece burs 2.35
Autoclaveable up to 135 degree
Nice design and good hand feeling
Fit ISO standard E-connect
Ball bearing type
|Set Included||Straight, Air Motor, Contra Angle|
|Working Air Pressure||0.22-0.3Mpa|
|Cooling Water||external water spray|
|Available Connection||universal dental unit connection|
Professionally used by dentists as oral operative surgery tools.
Notes & Cautions
1. Notes before operation
The product is available for dental inner spray air/electric motor in accordance with E type connector only.
Only to be used by a qualified dental practitioner.
Before using, please check burs and function of the chuck at outside of oral cavity of patients, if loose, too much vibration, irregular noise and too much heating up etc, please dot not carry on.
Avoiding crash and drop.
Forbidding disassembling or conversion.
It shall be sterilized before handpiece is used at the first time and after treatment.
Do not insert or remove bur while the bur is in motion.
Please use standard bur or drill.
Please do not exceed the max rotation speed of the bur or drill.
Please keep the bur or drill be clean. Any other foreign particles get inside, it may cause vibration, weak chucking power or other abnormalities.
2. Notes in operation
Please do not operate without spray.
Please do not let the motor speed exceed the max diving motor speed, otherwise will accelerate mechanical wear or damage the handpiece.
Please do not operate without bur or drill.
Please do not push the back cap while the handpiece is in motion, that will cause the handpiece damage
or the bur brust out.
3. Notes after operation
The contra angle is forbidden to put into digital ultrasonic cleaner, otherwise will damage the generator and LED light.
Please cleaning, lubricating and autoclaving after treatment. Otherwise the blood will make the cartridge corrosion.
The handpiece will heat up if lack lubricating.
Please do not use the acid solution to scrub, inject or soak the handpiece.
Please check the function before long time not use.
What are we supplying?
1. Dental unit
2.Air compressor (32L,38L,60L)
4. Sterilizing Equipments (Class B/N: 12L,16L,18L, 23L)
5. And many othe dental products.
Why choose use?
1. Complete Price List for wholesales
2, Supporting OEM Services for dental unit, air compressor, dental handpiece
3. Provide Professional After-Sale Services for customers
Mobile/WhatsApp/Wechat: 0086 13663715484
Contact Person: admin