|Place of Origin:||China|
|Minimum Order Quantity:||1pc|
|Delivery Time:||1-3days after got payment|
|Payment Terms:||L/C, D/A, D/P, T/T, , MoneyGram|
|Screen:||Colorful||Power:||100~250 (VAC), 50/60 (Hz)|
|Working Temperature:||0 ~ 40 Degree||Altitude Working:||-500 To 4,600m(-1,600 To 15,000ft)|
Surgery Multiparameter Patient Monitor,
Dental Multiparameter Patient Monitor,
dental pulse oximeter parameters
Colorful Portable Dental Implant Abutment Multiparameter Surgery Patient Moniter
Portable Patient Monitor (Figure 1-1) is adaptable to adult, pediatric and neonatal usage. It can monitor
vital signals as ECG, Respiratory Rate, SpO2, NIBP, TEMP and IBP. It integrates parameter measuring
modules, display and recorder in one device, featuring in compactness, lightweight and portability.
Replaceable built-in battery facilitates transportation of patient. Large high-resolution display provides
clear view of 5 waveforms and full monitoring parameters.
|Advantage||ECG, Respiratory Rate, SpO2, NIBP, TEMP and IBP|
|Transport and Storage Altitude||-500 to 13,100m(-1,600 to 43,000ft)|
|Transport and Storage Humidity||<= 93 %|
|Transport and Storage Temperature||-20 ~ 60 (C)|
The POWER switch is on the left quarter of the front panel (in Figure 1-1). The POWER indicator(in
Figure 1-1) and the BATT indicator ( in Figure 1-1) lights when the device is powered on. The
ALARM indicator flashes or lights when alarm occurs ( in Figure 1-1).The sockets of the sensors are
at the right side. The recorder socket is at the left side. Other sockets and power plug-in are at the
The system can display the alarm limit. The method is:
Select“SELECTION" item in“SYSTEM MENU" to access “SELECTION" sub-menu, in which the
user may set up the alarm limit. Set“ALM LIMIT" to ON to display the alarm limits of the paramneters
displayed on the screen or OFF to hide the alam limits.
Length of Alarm Records
The system may record the information prior to and after the occurring of alarm if physiological
alarm occurs. Three recording time is provided: 8s, 16s and 32s, which are the total length of the
time prior to and after the alarm. For example, 8s contains the respective information of 4s before
and after the alarm. 16s contains the respective information of 8s before and after the alarm, etc.
The user may select different recording time based on clinical requirement. The method is listed
Select“AL ARM SETUP" in“MONITOR SETUP" to access the sub-menu of“AL ARM SETUP".
In the“AI ARM REC TIME”item, the user may choose the length of alarm record. There are three
options for user to select 8s, 16s or 32s.
Select“TIME SETUP" iterm in“MONITOR SETUP" menu to access the sub-menu of “TIME
SETUP" as shown below. System time is in format of year, month, day, hour, minute and second Pick
the item you wish to modify and tum the knob, the figure will increase or decrease by 1 at each switch.
Then select“EXIT" item to retum to the previous memu.
New patient enrolment
Trend and Alarm Review
Portable Patient Monitor features flexible configurations. You can configure various aspects of the
monitor, inc1uding the parameters to be monitored, sweeping speed of the waveforms, audio signa1
volume, and printout text.
Press the“MENU"" hot key on the lower right part of the screen to call up“SYSTEM MENU". The
configuration is realized through operations on the SYSTEM MENU, as shown below.
Contact Person: Daisy Dai