|Place of Origin:||China|
|Certification:||CE, ISO, FDA|
|Minimum Order Quantity:||5pcs|
|Packaging Details:||5pcs/carton, 60*40*33cm, Gross Weight: 14KG|
|Delivery Time:||3-7 working days|
|Payment Terms:||TT, Western Union, Money Gram, RMB, D/P, D/A, L/C|
|Supply Ability:||3000pcs per month|
|Hanpiece:||Detachable||Compatible:||EMS And UDS|
|Function:||Scaling And Perio||MOQ:||4|
UDS Piezo Ultrasonic Scaler,
Piezo Ultrasonic Scaler,
UDS piezoelectric ultrasonic scaler
The product is used for teeth cleaning and also an important device for teeth disease prevention and treatment.
The UItrasonic scaler is composed of main unit, handpiece, cable, water pipe, tip, torque wrench, foot switch, and power supply.
|Power source||220V±10% / 50Hz ± 1Hz 150mA|
|Main unit input||24V 50Hz ± 1Hz 1.3A|
|Tips||G1*2, G2, G4, P1
Function and Operation:
3.1 Turn on the power switch, the power indicator lighted and the machine is ready for work.
3.2 Choose the scaling tip according to the requirement, and fix the scaling tip with the wrench. ( Please select a suitable power when using different type of tips (refer to "TABLE OF OPERATING POWER OF THE TIPS")
3.3 The handpiece can be handled in the same gesture as a pen in hand.
3.4 Vibrating intensity: Adjust the vibrating intensity according to your need, usually adjust to the middle grade, and adjust the vibrating during the clinical treatment according to the patient's sensitivity and the rigidity of the tartar.
3.5 Step on the foot switch, the tip begins to vibrate, and the LED lamp (model with LED) on the top of the handpiece light up elease the foot switch, the LED lamp keep shining for 10 seconds.
3.6 Under normal working condition, the frequency of the tips is very high, light touch and a certain to-and-fro motion will eliminate the tartar without obvious heating, overexetion and overstay are forbidden.
3.7 Water volume adjustment: Step on the foot switch, and the tip begins to vibrate, then turn the water control switch to fine spray to cool down the handpiece and clean the teeth
3.8 After finishing operation, keep the machine working for 30 seconds with the water supply to clean the handpiece and the tip.
3.9 Unscrew the scaling tip and sterilize it.
1.1.1Detachable handpeice can be autoclaved under 134 °C and 0.22 Mpa
1.1.2Automatic frequency tracking ensures that the device always works on the best frequency, stable and efficient performance.
1) The hemophilia patient is forbidden to use this equipment.
2) The patients or doctors with heart pacemaker are forbidden to use this equipmen
3) The heart disease patient, pregnant woman and children should be cautious to use the equipment
1.3 Equipment safety classification
1) Operating mode: Continuous operation
2) Type of protection against electric shock: Class II
3) Degree of protection against electric shock: Type B applied par
4) Applied part of the equipment: Tip
5) Degree of protection against harmful ingress of water: Ordinary equipment6) Degree of protection against harmful ingress of water: protection degree against water (used on foot switch):IPX17)
Degree of safety of application in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide Equipment can not be used in the presence of a flammable anesthetic mixture with air or with oxygen or nitrous oxide
1) Environment temperature: +5° C to +40° c
2) Relative humidity: 30%~75%.
3) Atmosphere pressure: 70kPa to 106kPa4) A temperature of the water at the inlet: not higher than +25
Cleaning, disinfecting and sterilizing:
5.1 Please conform to the recommendations of the manual "Reprocessing Instructions of Cleaning, Disinfecting And Sterilizing" (Attachment 2. )delivered with your product regarding the procedure of cleaning, disinfecting, sterilizing and packing of the components. Follow present-day regulations enforced in the country about reprocessing. The use of scouring powder or an abrasive sponge will damage its surface.5.2 Cleaning the accessoriesPlease conform to the recommendations of the manual "Reprocessing Instructions of Cleaning, Disinfecting And Sterilizing" delivered with your product. Follow present-day regulations enforced in the country about reprocessing. The handpiece, scaling tips, endochuck, torque wrench, endo wrench, LED lamp and Light conductor (the handpiece without LED lamp) can be sterilized. The cable and water pipe cannot be sterilized.
Contact Person: Echo Chai